FLUKE 5500A MANUAL PDF
Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
|Published (Last):||11 October 2007|
|PDF File Size:||13.66 Mb|
|ePub File Size:||15.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
Enter into the A the measured values when prompted. Readjust A90R13 if necessary to keep the edge signal at 10 ns to be at the reference level.
Page The memory holds flue constants and setup parameters. By pressing ‘print’ button you will print only current page. For example, a reading of 1.
Fluke 5500A/COIL Calculator User Manual
R is a square, single-turn pot and is marked on the board located near Q Vertical deflection characteristics are verified manjal calibrating the voltage gain.
Leveled Sine Wave flatness is measured relative to 50 kHz.
On the desired analog module, release the board edge locking ears. Remove all the analog modules.
A Operator Manual
DCV oltage V erific ation. Zeroing is particularly important when your calibration workload has 1 mil and 1 mV resolution, and when there are significant temperature changes in fluks A Calibrator work environment. Complete the pulse width calibration procedure. Connect the thermocouple to the front panel TC connector.
Remove the eight Phillips screws from the guard box cover. Volt Specifications A Operator Manual For capacitance outputs, null out stray capacitance by connecting the test leads to the UUT, routing them but not connecting to the A Calibrator on a non- conductive surface.
Flhke Summary by Function DC Volts Calibration Steps.
This warranty extends only to the original buyer or end-user customer of a Fluke authorized reseller, and does not apply to fuses, disposable batteries or to any product which, in Fluke’s opinion, has been misused, altered, neglected or damaged by accident or abnormal conditions of operation or handling.
To access calibration steps described in paragraphs throughsimply send the command: SC Hardware Adjustments Hardware adjustments must be made to the leveled sine and edge functions each time the SC is repaired. Figure shows a block diagram of the SC Option, also referred to as the A50 board. Edge frequency, duty cycle, rise time Procedure provided in this manual.
You have several choices for sending commands to the A: Watts uncertainty power voltage Sweeping Through a Frequency Range Entering Scope Cal mode prior to having the SC enabled for at least 5 minutes will cause a warning message to be displayed. Lift off the guard box cover using the finger pull on the rear edge of the cover.
FLUKE A OPERATOR’S MANUAL Pdf Download.
Maintenance Performing a Calibration Check Replacement Fuses Maintenance Introduction Returns the first error code contained in the A error queue, then removes that error code from the queue. Power and Dual Output Limit Specifications. Volt Specifications Table Front Panel Operation Setting the Output Program the Calibrator Mainframe to output as listed in Table High Frequency Flatness Verification at 25 mV.
Observe the A Calibrator Control Display changes back to the reset condition below. Ptft 0. Diagnostics runs 55000a following tests: Parts price information is available from the Fluke Corporation or its representatives. Verify the annunciator is on.
Program the Calibrator Mainframe to output the voltage listed in Table For this test, use the Fluke A, and an rms-responding meter. Page Operator Manual Serial Connectors The two 9-pin serial connectors on the rear panel of the A Calibrator are used to interface manuak a computer, or controller, and an instrument serial port. Input Impedance Mode Resistance. Adjusting the Edge Aberrations for Board A Enter 10 MHz into the Calibrator Mainframe. Remove the rear panel.
DC voltage references developed for the A. Set the Calibrator Mainframe MeasZ capacitance range to cap.